Receivers for vacuum shall meet the following requirements:
- Be made of materials approved by the manufacturer.
- Comply with Section VIII of the ASME Boiler and Pressure Vessel Code.
- Withstand a gauge pressure of 60 psi (414 k Pa) and 30 inch gauge HgV (102 kPa).
- Be equipped with a manual drain.
- Be of a capacity based on the technology of the pumps. [NFPA 99:126.96.36.199.3]